DECRETO LEGISLATIVO 1150 PNP PDF
Descripción: manual electrocardiógrafo Cardiofax DECRETO LEGISLATIVO N° DEL REGIMEN DICIPLINARIO DE LA PNP (RESUMIDO). docx. D.L. DECRETO LEGISLATIVO QUE REGULA EL REGÍMEN DISCIPLINARIO DE LA PNP TITULO PRELIMINAR PRINCIPIOS. Grupo Ascenso PNP. LEY DEL RÉGIMEN DISCIPLINARIO DE LA PNP – DECRETO LEGISLATIVO Nº DEL 11 DE DICIEMBRE DE
|Published (Last):||12 July 2015|
|PDF File Size:||7.40 Mb|
|ePub File Size:||18.11 Mb|
|Price:||Free* [*Free Regsitration Required]|
If a disposable electrode is used more than once, the ECG waveform cannot be recorded due to polarization voltage. Do not charge a deteriorated battery pack.
When equipotential grounding is required, always perform equipotential grounding by using the optional grounding lead. If you leave electrolyte cream cardioCream on the patient cable surface and leave the tip wet, it may cause inaccurate ECG recording due to high skin electrode contact impedance.
This is used to check deceeto recorder by printing a test pattern. Otherwise, the card might not operate. Examples of color-developing materials containing organic solvents: Settings Standard 12 leads: Correctly set the recording paper. Battery operation pbp Lights in lrgislativo Lights in orange Blinks in orange Available recording time At least 15 minutes. This is used to demonstrate the electrocardiograph.
Shipping costs must be pre-paid. Legislatuvo battery may leak. When installing or storing the instrument, take the following precautions: There is no setting file. To safely and effectively use the instrument, its operation must be fully understood. For inspection and replacement of the above parts, consult your Nihon Kohden representative.
Adjust the brightness with or If the screen is too dim, press the press the function key.
Manual Cardiofax 1150
Large Polarization Voltage If the message is still displayed after reattaching the electrode, it is probably due to a large polarization voltage.
Date and time Legispativo the date and time. Otherwise, water gets inside the electrocardiograph and causes malfunction. Failure to follow this warning may cause skin burn on the patient.
Marenbonearly Medieval Philosophy an Introduction. Electrodes with a deteriorated surface may cause high contact impedance decreyo distorted ECG waveforms. The customer or the user of the ECG can help prevent electromagnetic interference by maintaining a minimum distance between portable and mobile RF communications equipment transmitters and the Legiislativo as recommended below, according to the maximum output power of the transmitter.
Otherwise sudden power off occurs when an electrode is detached during recording. Set the recording paper.
decreto legislativo pdf – PDF Files
Measurement with excessive sensitivity: Immediately after electrodes are attached, the ECG waveforms may be unstable because of polarization voltage. During recording you can annotate the ECG waveforms when there is a change in patient condition such as chest pain or artifact.
You can make one or more copies of the recording results before the next recording starts. The list includes measurement date, ID, gender, age and measurement mode. Non-medical equipment in compliance with relevant IEC safety standards. There is a time delay between the input signal and output signal.
decreto legislativo 1150 pdf
Pull the card out straight from the SD card slot. Only use new electrodes.
Auto Gain and Auto Position The recording sensitivity and position are automatically adjusted so that the ECG waveforms of each channel fecreto not overlap.
When unexpected artifact appears, inspect the surrounding electromagnetic conditions and remove this artifact source.
Internal Memory Saves up to 40 files in the memory regardless of the data compression type. Feeds the paper while the key is pressed.
The electrocardiograph cannot manage data that is saved by another electrocardiograph.