EN ISO 6507-1 PDF
hardness test —. Part 1: Test method. Matériaux métalliques — Essai de dureté Vickers —. Partie 1: Méthode d’essai. STANDARD. ISO. Fourth edition. Purchase your copy of BS EN ISO as a PDF download or hard copy directly from the official BSI Shop. All BSI British Standards. I.S. EN ISO is the adopted Irish version of the European . The text of ISO has been approved by CEN as EN ISO.
|Published (Last):||10 August 2009|
|PDF File Size:||17.26 Mb|
|ePub File Size:||7.49 Mb|
|Price:||Free* [*Free Regsitration Required]|
Independent of the type of sources, for hardness there are two possibilities for the determination of the uncertainty. Werkstofftechnik 4pp. For automated measuring systems this standard should be applied accordingly. Unless otherwise specified, no part of this publication may be reproduced or utilized in any form or by any means, electronic or mechanical, including photocopying and microfilm, without permission in writing from either ISO at the address below or ISO’s member body in the country of the requester.
In accordance with Adobe’s licensing policy, this file may be printed or viewed but shall not be edited unless the typefaces which are embedded are licensed to and installed on the computer performing the editing. The support surfaces shall be clean and free from foreign matter scale, oil, dirt, etc. It should be noted that for anisotropic materials, for example those which have been heavily cold-worked, there will be a difference between the lengths of the two diagonals of the indentation.
VDI-Berichte 11pp. A guideline for the determination is given in Annex D.
Reference number ISO Your reading intentions are private to you and will not be shown to other users. It makes it easy to scan through your lists and keep track of progress. It may not always be possible to quantify all the identified contributions to the uncertainty.
The error b step 10 can be expected to be a systematic effect. Every care has been taken to ensure that the file is suitable for use by ISO member bodies.
ISO – Metallic materials — Vickers hardness test — Part 1: Test method
It is important that the test piece lies firmly on the support so that displacement cannot occur during the test. NOTE 2 In general, decreasing the test force increases the scatter of results of the measurements. The check involves at least one indentation being made on a hardness reference block, calibrated in accordance with ISO In downloading this file, parties accept therein the responsibility of not infringing Adobe’s licensing policy.
If the difference is greater, this shall be stated in the test report.
BS EN ISO 6507-1:2018
For undated references, the latest edition of the referenced document including any amendments applies. If the measuring system fails this test, appropriate action should be taken.
In the unlikely event that a problem relating to it is found, please inform the Central Secretariat at the address given below. Here’s an example of what they look like:. Naturally, direct calibration and the verification of these machines should be at the highest possible accuracy.
This iao particularly true for low-force Vickers hardness tests and Vickers microhardness tests, where the principal limitation will arise in the measurement of the diagonals of the indentation.
Test method 1 Scope This part of ISO specifies the Vickers hardness test method, for the three different ranges of test force for metallic materials see Table 1. The main task of technical committees is to prepare International Standards. The specification for the product may indicate 6507-11 for the differences between the lengths ne the two diagonals.
BS EN ISO – Metallic materials. Vickers hardness test. Test method
If not, an indirect verification should be 650-71. As a reference, an EA guideline  is available. Due to the small depth of Rn microhardness indentations, it is essential that special 65077-1 are taken during preparation. The time from the initial application of the force until the full test force is reached shall not be less than 2 s nor greater than 8 s.
Document Status Indicators The Green document status indicator indicates that the document is: The Amber document status indicator indicates that some caution is needed when using this document – it is either: Free to use BIM project management tool provides step-by-step help to define, manage and validate responsibility for information development and delivery at each stage of the asset life cycle in level 2 BIM projects.
Setting a reading intention helps you 65007-1 your reading. ISO consists of the following parts, under the general title Metallic materials — Vickers hardness test: The distance between the centres of two adjacent indentations shall be at least three times the mean diagonal length of the indentation in the case of steel, copper and copper alloys, and at least six times the mean diagonal length in the case of light metals, lead and tin and their alloys.
Details of 5607-1 software products used to create this PDF file can be found in the General Info relative to the file; the PDF-creation parameters were optimized for printing. Have you read this? These performance uncertainties reflect the combined effect to all the separate uncertainties indirect verification.
Figure 1 — Principle of the test The Vickers hardness is proportional to the quotient obtained by dividing the test force by the sloping area of the indentation, which is assumed to be a right pyramid with a square base, and having at the vertex the same angle as the indenter. Supersedes BS EN The Red document status indicator indicates that the document is an old version The document has likely been withdrawn by 6057-1 publisher, also the meta data presented here may be out of date as it is no longer being maintained by the editorial teams at NBS.
The correction factors are tabulated in terms of the ratio of the mean diagonal d of the indentation to the diameter D of the cylinder.