HP 438A MANUAL PDF
The A is a sophisticated microprocessor controlled dual channel power meter which can measure ratio and A HP Manual was downloaded from the Web. HP A Power Meter is a dual channel power meter designed specifically for ATE systems. The compact front panel is designed to save space in rack mounted. The HP A Power Meter is a dual channel power meter designed specifically for ATE systems. . Option Additional manual. Option W Two additional.
|Published (Last):||8 December 2014|
|PDF File Size:||19.70 Mb|
|ePub File Size:||17.48 Mb|
|Price:||Free* [*Free Regsitration Required]|
The power reference output and calibration controls are used to calibrate the particular power sensor in use.
Bottom view with cover removed. January A Quick Calibration Procedure: Disconnect Sensor from Power Ref output. Switch off Power Ref, check Oscillator light is off. Up to 19 different operating states of the meter can be stored into non-volatile memory for later recall.
Important measurement contributions are a programmable digital filter for optimizing resolution and measurement speed, independent offset in dB values for each channel, 0. November Checking all the specialist batteries in my test equipment, see Batteries In Test Equipment.
A HP Manual was downloaded from the Web.
Complete interrupt capability with flexible SRQ operation optimizes the efficiency of program execution in automatic systems. Top view with up removed. Compatible with the HP series of thermocouple and diode sensors, the power and frequency range of the meter extends from dBm to 44 dBm and from kHz to 50 GHz. The A has a 3V 0. Up to 19 different operating states of the meter can be stored into non-volatile memory for later recall. Initial power up display. The compact front panel is designed to save space in rack mounted systems, while the dual channel design allows simple measurements of the ratio and difference of power levels from two separate sensors.
The programmable digital averaging filter gives the user control over the inherent tradeoff between speed and accuracy. Welcome to Azur Electronics.
Keysight / Agilent A – In Stock, We Buy Sell Repair, Price Quote
I use this as my bp power meter. All measurement modes are programmable including zeroing, calibration, and Cal Factor. Rear Panel Sensor Connector and additional reference oscillator with rear panel output. Less digital averaging leads to faster but noisier readings if speed is the critical issue. The AUTO filter mode is usually adequate for fully settled readings with 0. The A needs powering up occasionally to recharge the battery.
HP A Power Meter
It measures absolute power in Watts or dBm, and relative power in dB. Initial display lights all segments.
View Cart My Account. Recommended for use with the Series Power Sensors. The A is a sophisticated microprocessor controlled dual channel power meter which can measure ratio and difference of power levels from 2 different sources. Display with no sensors connected.
HP 438A Manuals
Top PCB hinges up. Press Zero key and wait for routine.
The battery maintains the supply voltage to the RAM.